EOS recently introduced the innovative SPES2 technology with the aim of improving the information retrieved from each measured particle. The introduction of proper additional measurements further improves the EOS multiparamteric approach. EOS developed this approach through the SPES2 patented technology since 2017, basing on previous implementations realized within the EOS laboratories. The additional multiparametric approach is of utmost importance in characterizing samples with several unknown properties, such as the cases of strongly heterogeneous suspensions.
The SPES2 technology has currently been introduced to develop the EOS instrument aimed at characterizing airborne dust and/or micro- and nano-particles. The power scattered at 90° is collected and measured sinchronously to the traditional SPES signals. The corresponding adimensional parameter F90° is recovered thanks to the self-reference monitoring provided by the SPES sensor. This quantity provides the information corresponding to that obtained by the traditional OPC devices exploited for airborne particle measurements and for traditional particle sizer exploited for wet sample analysis.
An example of the superior insight allowed by a SPES2 device is reported in figure for airborne particles. The top panel shows the traditional SPES data. The additional powerful SPES2 data set is represented in the bottom panel. The forward scattering parameter, F0°, is just obtained from the SPES data and represented in greytones, while the additional F90° histogram is shown in blue. As it is clear, a bifurcation is evident in the F90° population, while it is completely hidden in the SPES data, both representations. A discrepancy of a decade is there in the F90° data for several fixed abscissas values. Moreover, we stress that the same bifurcation would be completely invisible to a traditional instrument as Optical Particle Counter OPC, operating with the one F90° parameter. This bifurcation shows a clear twofold population with rampand different optical properties.
The SPES2 technology has been recently exploited for developing an instrument devoted to characetrize the optical properties of airborn dust in the environement, with a current application within the OPTAIR research project funded by the Italian National Research Antarctic Program. The instrument is continuously working since november 2018 at Concordia Station, DomeC, on the East Antarctic plateau.
(in figure) The traditional SPES data (top) and the SPES2 corresponding population (botom), constituted by the adimensional forward scattering amplitude (black), obtained form the left panel data, and the additional F90° scattered adimensional intensity (blue). Notice the presence of a bifurcation in the F90° population.